Get HYDAS User manual. Get all HYDAS manuals!. Instruction Manual. TENS device – «Easy Free» TENS. Please read before using! GB. GAz1_ Hydas GmbH & Co. KG | Hirzenhainer Straße 3 | Hydas Easy Free Hair Removal System () – now buy online with ease from , your Easy Free; Tweezers; Power supply; Operating instructions.
|Published (Last):||18 August 2006|
|PDF File Size:||18.39 Mb|
|ePub File Size:||3.75 Mb|
|Price:||Free* [*Free Regsitration Required]|
Hydas AD-2011 Manuals
Are the batteries properly loaded? Have the electrode pads come Turn off the power and stick off the skin? Are the batteries exhausted? Firmly connect the wire. Turn off the power and properly connect the wire. BoxHershey, PA. These are normally transmitted from the area of damage through the nerve fibers to the brain, TENS interrupts these pain signals.
Page 13 Power sources cut off in the therapeutic process. Maual of therapy GB Transcutaneous electrical nerve stimulation TENS is a non-invasive technique in which a low-voltage electrical current is delivered through wires from a small power unit to electrodes located on the skin.
The customer or the user of the should assure that it is used in such an environment. Ar tike l Ar tico lo Ar ticulo. Tell us about it. Don’t see a manual you are looking for? The time display is enabled and displays the remaining time of application. Der Kunde oder der Benutzer des Artikels hat sich zu vergewissern, dass er in einer solchen Umgebung eingesetzt wird. Page 14 Environment Protection – Waste electrical products should not be disposed of with household waste. Environment Protection – Waste electrical products should not be disposed of with household waste.
GB Number of treatment program: Page 6 GB Number of treatment program: Page 11 8 Operating instructions a Be sure you know the right points where to place the electrodes and affix them there then, connect the electrodes to Digital Pain Relief with wire. Correctly load the batteries.
Is the wire properly connected? Tell us what’s missing. Replacement pads are available from your local distributor. Please gydas not pull the wire.
If the measured field strength in the location in which the is used exceeds the applicable RF compliance level above, the should be observed to verify normal operation. To assess the electromagnetic environment due to fixed RF transmitters, an electromagnetic site survey should be considered. Malfunction The reason Countermeasure You have no feeling of stimulus. Check with your local Authority or retailer for recycling advice. Or point us to the URL where the manual is located.
Power sources cut off in the therapeutic process. The skin becomes red. Please recycle where facilities exist. Ar tike l Ite m Ar ticl e.
Hydas N/A User Manual – Page 17 of 17 |
bydas Your manual failed to upload Electrodes are temporarily attached with paste in various patterns, depending on the specific condition and treatment goals. Please replace them with new ones. This may cause burn injuries hycas the skin under the electrodes, as well as problems with the stimulator. If abnormal performance is observed, additional measures may be necessary, such as re-orienting or relocating the Finishing the operation You should hold the plug when pulling it out.
Are the wires disconnected? Have the batteries been exhausted? Page 22 and TV broadcast cannot be predicted theoretically with accuracy.
Ple ase r ead hydqs ef ore u sing! TENS is often used to treat pain, as an alternative or addition to pain medications. Do not immerse the device in water.